We can measure hardness of material by Vickers method using UMT Tribolab machine. This is a tutorial step by step to test hardness of glass.
Make a indentation Vicker with condition:
- Load: 2N
- Time: 15s
- Sensor: DFM 2G (maximum 20 N)
Make new script: click script tab>make consequence>add step
Engage tab is a selection touch force (N). It means that sensor will see to tourch surface of material with force equals 0.05 N.
Data file: select location is saved data
Step 1 : make a load is 2 N in time is 15s: Select General tab and chanel is Fz, mode is Constant Load, Duration is 15 seconds.
Load control tab Mode is Constant Load, Fz value -2N
Linear Drive Mode is IDLE
Sider tab Mode is IDLE.
Step 2: offset tip up 0.3 mm: General time is 3s, Cariage tab mode is relative, offset is 0.3 mm, Velosity is automatic
Linear Drive and Sider is selected mode Idle
Step 3: Move tip Vicker go to right 0.2 mm. Our target is make second indentation vicker
Distance is 0.2 mm, velocity is 0.5 mm/s. Total time is 6s
Indentation 2,3,4,5 like indentation 1
After make indentation Vicker, We measure d1 and d2 by SEM machine:
And then using tool UMT TRIBOLAB machine or Vicker fomular https://en.wikipedia.org/wiki/Vickers_hardness_test
To caculate value of Vicker hardness Kgf/mm2 or GPa
This is check hardness for glass method example. Besides, we can select other suiable conditon for other material.
key: #vntoday#vntoday.com.vn#measure hardness#measure hardness material#measure hardness of material 2018#Measure hardness material 2018#using UMT Tribolab#practice check hardness material#hardness material#SEM#SEM machine#hardness of glass#hardness of glass 2018